Offloading Difficult Wounds and Conditions in the Diabetic Patient
- See this Informative and well Referenced Technical Article on Heel Pressure Ulcer Treatment with specific mention of KYDEX-PRO LEEDer Boots-CLICK-HERE
Offloading Difficult Wounds and Conditions in the Diabetic Patient
- Pressure reduction is the cornerstone to preventing and healing foot ulcers in people with neuropathy. Total contact casting (TCC) is the most commonly used and extensively researched method to reduce plantar pressure.
- However, TCC is not the ideal treatment for all patients with wounds and rarely used for prevention.
- Fortunately, today clinicians may choose from many different offloading devices. Because their optimal use and effectiveness often hinges upon comfort and ease of use, a careful review of all options and their advantages and disadvantages can, literally, save a limb.
Issue Number: 1 Author: Robert J. Snyder, DPM, FACFAS, CWS, and Karen K. Lanier, CPed
- Lower extremity wounds represent significant medical and financial challenges to the healthcare system. This is especially the case in the diabetic population where neuropathy, often associated with vascular disease, can lead to ulceration, immobility, infection, and gangrene. One study places the average cost of healing a single diabetic ulcer at approximately $36,000,1 leading to direct costs of $600 million dollars annually.2 Limitations on mobility caused by foot ulcers have a negative impact on social, psychological, physical, and economic domains.3 Charcot arthropathy can create additional complications; all of these sequelae can lead to major limb amputation.1 Early recognition and prompt management of these foot ulcers can facilitate healing.4 Additional data suggest that meticulous, scientifically based wound care and patient education strategies can reduce lower extremity amputation rates by reducing the frequency and severity of foot ulcers.5
- In response to the high costs of managing chronic wounds, systems of comprehensive outpatient care have been established and represent an interdisciplinary and collaborative approach to wound management. The goals of wound care protocols are to treat underlying conditions that cause wounds, facilitate the wound healing process, and minimize skin breakdown and wound recurrence.6 The scope of potential interventions includes debridement, infection control, offloading, protective and active dressings, revascularization, proper nutrition, and patient education.7 Offloading is important for reducing foot pressure points8 and for prevention,9 as well as for healing.10
The following article discusses offloading difficult wounds and Charcot foot at various stages of treatment and prevention. The pedorthic management of foot amputations also will be discussed.
The Neuropathic Ulcer
- The main cause of foot ulceration is neuropathy (see Figure 1).11 This leads to prolonged and excessive pressures that cause tissue breakdown. Using footwear as a means of healing open wounds is rarely desirable.12 Currently, total contact casting (TCC) (see Figure 2) represents the gold standard for the treatment of forefoot and midfoot (Wagner grade 1-2) diabetic and neuropathic ulcerations; however, reduction of heel pressures with this device remains controversial. Myerson et al13 estimate that 6 weeks of treatment in a total contact cast costs the same as a single day of inpatient treatment. Specialized casting protects the foot from trauma, immobilizing skin edges and reducing edema. It decreases pressure over the ulcer by redistributing the weightbearing load over a greater plantar surface area.
- Molding the bottom of the cast to the bottom of the foot causes the entire sole to participate in the force distribution, resulting in lower pressures.14 In 1985, Birke et al15 reported 75% to 84% reduction of peak pressure at the first and third metatarsal heads, respectively, when subjects walked in a cast. Many other studies have supported the successful rationale for TCC use. For example, Sinacore et al16 noted healing in 82% of 33 ulcers after an average of 44 days in a total contact cast. Myerson et al13 observed healing in 64 out of 71 (90%) ulcers at a mean of 5.5 weeks. Hanft et al17 performed a 10-year retrospective study utilizing total contact casts on more than 1,000 patients. The research demonstrated a healing rate of 91% within 13 weeks with an average closure time of 4.36 weeks (± 1.31 weeks). It was concluded that the total contact cast was an effective, low-risk, and inexpensive treatment for plantar diabetic foot ulcers.
- Total contact casting has several disadvantages, including joint stiffness, muscle atrophy, the possibility of new ulcerations and skin breakdown, labor-intensive application, and possible laceration of the patient during cast removal. Contact dermatitis and fungal infection also may occur and should be treated with appropriate topical medications and temporary abstinence of casting. Patient compliance due to the absence of pain is also an issue. Although TCC is an ambulatory procedure, the patient is required to limit ambulation to one-third normal. This often requires counseling and close follow-up while the cast is in place. Vascularity must be carefully evaluated before cast application. Lavery et al18 discovered that TCC causes postural instability in the ambulating patient as compared to a tennis shoe or removable cast walker; therefore, consideration for the well being and safety of the patient must be strongly considered before recommending the device.
- Alternatives to TCC may include the following:
- Nonweight bearing
- Standard below-knee casts
- Charcot Restraint Orthotic Walkers (CROW)/total contact brace
- Prefabricated walker
- The DH walker (Centec Orthopaedics, Camarillo, Calif.)
- The Integrated Prosthetic and Orthotic system (IPOS, Niagara Falls, NY)
- Orthowedge (Darco International, Huntington, WV)
- Healing sandal
- Reverse IPOS heel relief shoe system
- L?Nard Splint/MultiBoot? (AliMed Inc., Dedham, Mass.)
- Standard ankle/foot orthoses (AFO)
- Patella tendon bearing brace (PTBB)
- Prefabricated pneumatic walking brace (PPWB)
- MABAL shoe/Scotch boot
- Additional considerations.
Nonweight bearing. This can be accomplished with bed rest, wheelchair, walkers, crutches, and similar strategies and appliances. However, wheelchairs are cumbersome and may not permit mobility in the patient’s home. Crutches and walkers may actually increase pressure to the contralateral limb.19 Diabetics may lack the upper body strength, cardiovascular reserves, or motivation to consistently use these modalities.20 Noncompliance is a significant issue with these devices.
Standard below-knee casts. Several studies support similar efficacy between the total contact cast and the standard cast.21 Plaster or fiberglass tape appears to be equally efficacious.22 However, patients with neuropathy run the risk of injury from the edge of the cast or from material working its way into the open toe design; therefore, the closed toe design of the TCC appears to be superior.
Charcot Restraint Orthotic Walker (CROW). This device (see Figure 2) is also known as a ?clamshell? or bivalved AFO (BAFO). It is composed of a polypropylene material, lined with plastizote, incorporates a total contact custom molded orthotic, and utilizes a rocker bottom sole. The CROW is the treatment of choice for many clinicians during the quiescent second and third Eichenholtz stages of the neuropathic process where joint stability and alignment must be maintained.23
The advantages of the CROW include:
- ability to inspect and treat the ulcer as needed by removing the appliance
- minimal joint stiffness and atrophy
- ability to utilize growth factors and other topical medications and dressings
- ability to control edema
- allows for ambulation
- provides patient satisfaction.
Among the CROW’s disadvantages:
- removability – the patient can remove and ambulate without protection
- cumbersome relative to size and weight and may not be appropriate for frail individuals, patients with motor difficulties, or morbidly obese patients
- likelihood of skin irritation and breakdown – should be worn over a support stocking or knee-high cotton hose
- requires frequent adjustments to accommodate for reductions in edema.
Prefabricated walker. Originally designed for the treatment of fractures and sprains, a prefabricated walker may be useful in the offloading treatment of diabetic foot ulcers.24 Hanft et al25 performed a retrospective study of more than 300 patients with plantar diabetic foot ulcers and demonstrated a healing rate of 85% within 13 weeks with an average time to closure of 5.51 weeks (± 1.02 weeks). In another study, the prefabricated walker has been shown to reduce foot pressures by as much as 56% to 58% and one specific model by 63% to 70%.26 Diabetic ulcerations must be protectively padded and closely monitored. Prefabricated walkers are of additional benefit because they can be removed to inspect and treat ulcerations as needed and can be reused in cases of recidivism. Other advantages appear similar to that of the CROW boot. The caveat lies in the patient?s potential for noncompliance (see Figure 2).
The DH walker. This device is similar to the commercial walking brace; however, the floor of the device contains prongs made of polyethylene that can be removed to effectively accommodate an ulceration or deformity. These removable prongs (hexagons) are also available in an offloading shoe design. Fleischli et al26 postulated that the DH pressure relief walker reduced plantar pressures by 79% to 80% and appeared to be as efficacious as the total contact cast in reducing plantar pressures in the great toe. In this study, the device appeared to be superior to the TCC in reducing plantar pressures for ulcerations under the first metatarsal (see Figure 2).
Orthowedge. This device is similar to the IPOS design except the sole extends to the toes. Again, the patient’s ability to perform adequate dorsiflexion is at issue. One study showed plantar pressure reduction of 64% to 66% utilizing this modality, yet the device was less effective than the TCC or DH walker (see Figure 2).19
Healing sandal. This device is constructed with a dual density total contact orthotic made of plastizote. The orthotic contains a cutout under the ulcerated area, and the edges are skived (beveled) to minimize stress. The insole is then placed in a Darco-type postsurgical shoe. Hanft et al29 performed a study of more than 500 patients with plantar diabetic foot ulcerations and demonstrated a healing rate of 74% within 13 weeks with an average time to closure of 7.11 weeks (± 2.35 weeks). Advantages include a lightweight design that is more aesthetically pleasing than the TCC or CROW boot. The main disadvantage is poor control of foot motion, which may lead to increasing stress at the lesion site (see Figure 2).
Reverse IPOS heel relief shoe. This appliance offloads at the heel with a shoe design that is open at the back and angled in 10 degrees of plantar flexion. Although lightweight and relatively inexpensive, this device may create gait instability and difficulty with balance (see Figure 2).
L’Nard splint/multiboot. This device enables pressure-free suspension without foot irritation and features a swing-out antirotation bar and height-adjustable footplate to keep the bedding from pressing on the toes and foot (see Figure 2).
Ankle foot orthoses (AFO). This device is constructed of molded thermoplastic material and represents a rigid ankle design. Used with or without a dorsal foot strap, it is indicated for a multitude of conditions including drop foot, rotational control and weight reduction of a residual foot, and for medial to lateral stability. An AFO can be made in a rigid or articulating design (see Figures 3a and 3b).
Prefabricated pneumatic walking brace (PPWB). Using inflatable and adjustable air cells, this device functions as a removable cast. One study revealed that the PPWB was superior to the TCC in decreasing plantar pressures.32 Benefits include reasonable cost and the ability to inspect a wound. One disadvantage is the potential for patient noncompliance.
MABAL shoe/Scotch boot. This is a removable fiberglass combination of a cast and a shoe (the initials represent an acronym for its inventors and place of origin). Hissink et al33 studied patients with a total of 23 ulcerations and discovered that 21 healed with a mean healing time of 34 days (7 to 75 days). It was concluded that the MABAL shoe provided healing of neuropathic diabetic foot ulcers comparable to existing methods of treatment. The main advantages of this modality include mobilization of the ankle, removability, and less time-consuming application when compared to the TCC. An additional modification formulated by the authors includes the attachment of Velcro? between the bottom of the shoe and the sole, allowing for the removal of the sole during sleeping, if desired (see Figure 2).
Felt and foam total contact padding. This technique creates an aperture (or opening) in a piece of felt and/or foam to accommodate an ulcer. The material is subsequently applied directly to the full length of the plantar aspect of the foot. This pressure-relieving strategy is easy to apply, usually causes no secondary lesions, encourages patient compliance, and is cost effective. Unfortunately, much of the evidence regarding efficacy of this modality is anecdotal.34
Bledsoe conformer diabetic boot (Bledsoe Brace Systems, Medical Technology Inc, Grand Prairie, Tex.). This device was specifically designed for diabetic patients with problematic ulcerations. The boot features a fully enclosed thick foam cocoon attached to an auto-molding innersole that embeds into a specially designed premolded midsole insert in aluminum shelled walking boot. Pollo et al35 conducted a study on 10 normal subjects with no prior history of foot or ankle problems and determined that the Bledsoe conformer functioned as well or even better than a total contact cast with the added convenience of allowing removal of the boot for wound examination. However, this study represents a small sampling and additional investigation is underway to expand clinical data with regard to healing time and pressure differences.
Once the wound is healed, maintaining closure is imperative. Recurring ulceration is usually due to noncompliance with prescription shoe wear. A lack of protective shoe gear is the most common cause of ulcer recurrence.36 A study undertaken by Helm et al37 revealed a recurrence rate of 19.4% out of 102 patients. Myerson et al13 postulated a recurrence rate of 31% out of 71 patients. These findings were supported by a Centers for Disease Control study that determined that diabetics with proper shoe protection had only a 20% recurrence, while those without had an 80% recurrence.36
Preventing recidivism. The following modalities can be utilized to prevent recidivism.
Depth shoes. These shoes provide additional vertical depth of 3/16 to 1/2 inch. Wide shank depth shoes give additional width in the girth (instep).
Depth wide shank shoes. This shoe is 1/4-inch deeper than regular depth shoes and will accommodate up to a 1/2-inch inlay.
Full contact orthoses/total contact insert (TCI). Special insoles reduce foot pressures by as much as 50%.40 These devices are molded over a positive model or directly to the patient?s foot with appropriate posting. This modality is made of material that may accommodate the deformity, and/or alters lower extremity biomechanics. A full-contact orthosis requires shoes of sufficient depth.
Steel shanks and extended shanks. Steel bars are used with the rocker sole to reduce metatarsophalangeal extension or provide better biomechanics for the transmetatarsal amputee.
SACH heel and other heel modifications. The insertion of a SACH heel to the rear midsole of a patient’s shoe acts as an extrinsic shock absorber. This device is constructed of a high-density material with give and is inserted to prevent wear while absorbing shock and immediately rebounding for the next heel strike. SACH heels often are used in conjunction with short leg braces and can replace some of the lost rotational movement of the limb on the foot when such torque is lost due to fusion or trauma.41 This device also can stabilize the foot after partial calcanectomy. Wedging is frequently used for stabilizing a flexible deformity in a corrected position or in accommodating a fixed deformity. A medial wedge is indicated in extreme pronation, and a lateral wedge can be used for ankle instability or a varus heel deformity. Wedges can be placed between the upper and the sole or directly on the bottom of sole. Heel and sole flaring often are used for support. This type of modification provides medial lateral stability to the foot on the side that it is applied. Flaring is often united with wedging for maximal effect (see Figure 4).
Rocker bottoms. Rocker-sole shoes are one of the most commonly prescribed shoe modifications.42,43 They represent exterior additions to the outsole that taper off at the distal tip and may taper off to the posterior edge of the heel, creating changes in biomechanical function.
In general, rocker soles are custom made for each patient; six basic types of rocker soles can be identified based on the variation of position and the degree of the rocker angle (see Figure 5).
1. Mild rocker sole – The most widely used and most basic of the rocker soles has a mild angle at both the heel and the toe. This type of rocker sole can relieve metatarsal pressure and may assist gait by increasing propulsion and reducing energy expended when walking. It is appropriate for the foot that is not at risk and typically found on athletic walking shoes.
2. Heel-to-toe rocker sole – Shaped with a more severe angle at both the heel and the toe, the heel-to-toe rocker is extended to provide greater propulsion at toe off. This causes a decrease in heel strike and reduces the need for full range of ankle motion. The modality is appropriate for a fixed claw toe, rigid hammertoe, midfoot amputation, calcaneal ulcers, or for the patient who has undergone triple arthrodesis.
3. Toe-only rocker sole – The purpose of a toe-only rocker sole is to increase weight bearing behind the metatarsal heads, provide stability at midstance, and reduce the need for toe dorsiflexion. It is useful for hallux rigidus, hammertoes, and metatarsal ulcers associated with diabetes.
4. Severe angle rocker sole – This type of sole also has a rocker angle only at the toe; however, the angle is much more severe than that found on the toe-only rocker sole. This device eliminates the weight-bearing forces at the metatarsal heads and anterior to them and is most often used for extreme relief of diabetic ulcers at the metatarsal heads.
5. Negative heel rocker sole – Shaped with a rocker angle at the toe and a negative heel, this type of rocker sole results in the patient’s heel being at the same height or lower than the ball of the foot when the patient is standing. The negative heel rocker sole accommodates a foot that is fixed in dorsiflexion and/or relieves forefoot pressure by shifting it to the hind foot and midfoot. The negative heel rocker sole should be used with caution because the patient’s inability to attain the appropriate ankle dorsiflexion may cause discomfort and ultimately increase pressure on the problem area.
6. Double rocker sole – This is a mild rocker sole with a section removed in the midfoot area; thereby, giving the appearance of two rocker soles – one at the hind foot and one at the forefoot (two areas of midstance). Because the thinnest area of the double rocker sole is at the midfoot, it is used to relieve a specific midfoot problem area, such as the Charcot foot deformity associated with diabetes.
Charcot Foot (Osteoarthropathy)
The etiology of Charcot foot (see Figure 6) is multifactorial and based on neuropathy and metabolic processes.44 Neuropathic arthropathy may be defined as a relatively painless, progressive, and degenerative condition of single and multiple joints caused by underlying neurological deficits.45 Additional neurovascular theory postulates that increased peripheral blood flow due to autonomic neuropathy leads to a hyperemic bone resorption.46 Other predisposing factors may include renal transplantation, immunosuppressive therapy, impaired cartilage growth, and nonenzymatic glycosolation.45 Sadly, these deformities can result in major limb amputation.47
Eichenholtz48 described the stages of bone and joint destruction followed by fracture healing and remodeling. This temporal classification is based on the characteristic clinical and radiographic changes that occur with neuropathic joint destruction and fracture over time; progression occurs from the acute phase (dissolution) through the healing phase (coalescence) to the resolution phase. The resulting foot deformities cause difficulty with shoe fit and significantly increase the propensity toward ulceration in high-pressure areas.49 Almost all patients with these deformities will ultimately require specialized footwear with custom total-contact inserts or custom bracing. Surgery for these conditions is predicated on the goal that restoration of stability and alignment of the foot and ankle will make appropriate shoeing and bracing possible.50 Foot ulcers in association with significant neuropathic deformity are usually treated with many modalities, ranging from total-contact casts, CROW boots, AFO, PTB devices, and removable prefabricated walking braces to custom molded insoles, bracing, and shoes. The type of appliance used is predicated on the Eichenholtz stage of development.
Stage I (Dissolution). Radiologically, regional bone demineralization with periarticular fragmentation and joint disarticulation is present. Clinically, acute inflammation, swelling, erythema, and heat are evident. This early stage is easily mistaken for infection or thrombophlebitis.51 Treatment is nonweight-bearing and usually consists of a total contact cast, BAFO, CROW boot, or a removable prefabricated walking brace. One study of 34 feet revealed that treatment averaged 8.4 months. However, follow-up could continue for 35 months or more.52 As treatment progresses, care must be taken to gradually wean the patient from nonweight-bearing to partial and then full weight-bearing along with the help of assistive devices; these include appropriate and professionally guided use of crutches, walkers, or canes.53
Stage II (Coalescence). Radiologically, absorption of bone debris in the soft tissues is evident, along with organization and early healing of fracture fragments and periosteal new bone formation. Clinically, a decrease in inflammation with less fluctuance can be noted as well as increased stability of fracture segments. A study of 43 feet in this classification30 revealed that immobilization lasted an average of 6.4 months in 34 of the feet and follow-up averaged 16.4 months.
Stage III (Resolution). Radiologically, a smoothing of large bone fragment edges is present, along with sclerosis and bony/fibrous ankylosis. Clinically, permanent enlargement of the foot and ankle, fixed deformity, minimal swelling, and normalization of temperature are evident. A study of 60 feet in this category30 required follow-up for 22.1 months. Treatment consisted of custom-molded shoes and insoles unless ulceration dictated otherwise. No immobilization was required in this group because of the healed nature of the process. Anderson et al21 recommends an extra-depth shoe coupled with a cushioned insert attached to a double-upright calf-lacing brace with a fixed ankle joint and rigid rocker sole. The shoe may need a wide shank or a more rigid rocker if the midfoot or hindfoot areas are involved. The patient also should be fitted with knee-high compression hose. An AFO or CROW can be used as an alternative.
Pedorthic Management of Amputations of the Foot
Partial amputations of the foot are becoming more frequent when compared with transtibial or transfemoral amputations because retained length results in better function and lower energy expenditure.54,55 The goal of surgery is to achieve the most distal level of amputation that will heal. These procedures should result in a stable shoeable or braceable residual limb that allows for maximum rehabilitation.56 The prosthetist should strive to provide a comfortable, safe, stable, and supportive device that is easy to use and adjust.57,30 Function of the amputated part should be stable enough so as not to cause breakdown of the contralateral limb.
Table 1 represents the most common foot amputations and recommendations for accommodation, shoeing, and bracing.
1. Greene DA, Feldman EL, Stevens M. Neuropathy in the diabetic foot: new concepts in etiology and treatment. In: Levin ME, O?Neil LW, Bowker JH, eds. The Diabetic Foot, 5th ed. St. Louis, Mo.: Mosby Year Book; 1993. 2. Jiwa F. Diabetes in the 1990s. An overview. Stat Bull Metro Insurance Company. 1997;78:2-8. 3. Brod M. Quality of life in patients with diabetes and lower extremity ulcers: patients and caregivers. Qual Life Res. 1998;7:365-372. 4. American Diabetes Association. Position statement; foot care in patients with diabetes mellitus. Diabetes Care. 1995;18(suppl 1):S26-S27. 5. National Institutes of Health, National Institute of Diabetes and Digestive and Kidney Diseases (NIDDK). Prevention and early intervention for diabetes foot problems. Feet Can Last a Lifetime. Bethesda, Md.: NIDDK; 1998. 6. Glover JL, Weingarten MS, Buchbinder DS, et al. A 4-year outcome-based retrospective study of wound healing and limb salvage in patients with chronic wounds. Advances in Wound Care. 1997;10:33-38. 7. Steed DL, Donohoe D, Webster MW, et al. Effect of extensive debridement and treatment on the healing of diabetic foot ulcers. J Am Coll Surg. 1996;183:61-64. 8. Boulton AJM. The diabetic foot. Med Clin North Am. 1988;72(6):1513-1530. 9. Janisse DJ. Prescription insoles and footwear. Clin Pod Med Surg. 1995;12:41-61. 10. Lavery LA, Vela SA, Lavery DC, et al. Reducing dynamic foot pressures in high-risk diabetic subjects with foot ulcerations: a comparison of treatments. Diabetes Care. 1996;19:818-821. 11. Levin ME. Preventing amputations in the patient with diabetes. Diabetes Care. 1995;18:1383-1392. 12. Coleman WC, Brand PW, Birke JA. The total contact cast. J Am Podiatr Assoc. 1984;74:548. 13. Myerson M, Papa J, Eaton K, et al. Total contact casting for the management of neuropathic plantar ulceration of the foot. Amer J Bone Joint Surg. 1992;74A:261-269. 14. Dhawan S, Conti S. Use of total contact casting in the diabetic foot. Foot and Ankle Clinics. 1997;2(1):115-136. 15. Birke JA, Sims DA, Buford WL. Walking casts: effect of plantar foot pressures. J Rehabil Res Dev. 1985; 22:18. 16. Sinacore DR, Meuller MJ, Diamond JE, et al. Diabetic plantar ulcers treated with total contact casting: a clinical report. Phys Ther. 1987;67:1543-1549. 17. Hanft JR, Surprenant MS. Is total contact casting the gold standard for the treatment of diabetic foot ulcerations? Abstract presented at: Joint Annual Meeting and Scientific Seminar; February 9, 2000; Miami, Fla. 18. Lavery LA, Fleischli JG, Laughlin TJ, et al. Is postural instability exacerbated by offloading devices in high-risk diabetics with foot ulcers? Ostomy/Wound Management. 1998;44:26-34. 19. Armstrong DG, Liswood PL, Todd WF. The contra-lateral limb during total contact casting: a dynamic pressure and thermometric analysis. J Am Podiatr Med Assoc. 1995;85;733-737. 20. Armstrong DG, Lavery LA, Harkless LB. Options for off-loading the diabetic foot. Wounds. 2000;12(6 Suppl B):30B-34B. 21. Pollard JP, LeQuesne LP. Method of healing diabetic forefoot ulcers. Br Med J. 1983;286:436-437. 22. Huband MS, Carr JB. A simplified method of total contact casting for diabetic foot ulcers. Contemporary Orthopedics. 1993;26:143-147. 23. Anderson RB, Davis WH. The pedorthic and orthotic care of the diabetic foot. Foot and Ankle Clinics. 1997;2(1):137-151. 24. Ayaso F, Gorgon D, Lui E. Review of treatment modalities in the off-loading of diabetic foot ulcers. Podiatric Medical Review. 2000;6(2):55-59. 25. Hanft JR, Surprenant MS. The use of the fixed ankle walker for the treatment of plantar diabetic foot ulcerations. ACFAS Abstract presented at: Joint Annual Meeting and Scientific Seminar, American College of Foot and Ankle Surgeons; February 8-12, 2000; Miami, Fla. 26. Fleischli JG, Lavery LA, Vela SA, et al. Comparison of strategies for reducing pressure at the site of neuropathic ulcers. J Am Podiatr Assoc. 1997;87:466-472. 27. McDermott JE, ed. The Diabetic Foot. American Academy of Orthopedic Surgeons Monograph series. 1995;17-18. 28. Chantelau E, Breuer U, Leisch AC, Tanudjada T, et al. Outpatient treatment of unilateral diabetic foot ulcers with the ?half-shoes.? Diabet Med. 1993;10:267-270. 29. Hanft JR, Surprenant MS. The use of the custom molded healing sandal for the treatment of plantar diabetic foot ulcerations. ACFAS Abstract presented at: Joint Annual Meeting and Scientific Seminar, American College of Foot and Ankle Surgeons; February 8-12, 2000; Miami, Fla. 30. Rheinstein J, Yanke J, Marzano R. Developing an effective prescription for lower extremity prosthesis. Foot and Ankle Clinics of North America. 1999;4(1):113-138. 31. Guse ST, Alvine FG. Treatment of diabetic foot ulcers and Charcot neuroarthropathy using the patellar tendon-bearing brace. Foot and Ankle. 1997;18(10):675. 32. Baumhauer JF, Wervey R, McWilliams J, et al. A comparison study of plantar foot pressure in a standardized shoe, total contact cast, and prefabricated pneumatic walking brace. Foot Ankle Int. 1997;18:26-33. 33. Hissink RJ, Manning HA, Van Basal JG. The MABAL shoes, an alternative method in contact casting for the treatment of neuropathic diabetic foot ulcers. Foot Ankle Int. 2000;21(4):320-322. 34. Guzman B, Fisher G, Palladino SJ, Stavosky JW. Pressure-removing strategies in neuropathic ulcer therapy. Shoes, Orthoses, and Related Biomechanics, Clinics. Podiatric Medicine and Surgery. 1994;11(2);339-353. 35. Pollo FE, Brodsky JW, Crenshaw SJ, and Kirksky C. Plantar pressures in total contact casting versus a diabetic walking boot. www.bledsoebrace.com. Accessed 11/19/01. 36. Hayes S. The pedorthic prescription. Ambulatory Foot Care Course, American Academy of Orthopaedic Surgeons, San Francisco, Calif.; 1987. 37. Helm PA, Walker SC, Pullium GF. Recurrence of neuropathic ulcerations following healing in a total contact cast. Arch Phys Med Rehabil. 1991;72:967-970. 38. Centers for Disease Control, Disease Prevention and Health Promotion. Economic aspects of diabetes services and education. US Department of Health and Human Services, Atlanta, Ga. Selected annotations. 1992. 39. Lavery LA, Lavery DC, Quebedeaux-Farnham TL. Increased foot pressures after great toe amputation in diabetes. Diabetes Care. 1995;18:1460-1462. 40. Murray HJ, Boulton AJM. The pathophysiology of diabetic foot ulceration. Clin Podiatr Med Surg. 1995;12:41-61. 41. Frykberg RG, Kozak GP. The diabetic Charcot foot. In: Kozak GP, Hoar CS, Rowbotham JL, et al, eds. Management of Diabetic Foot Problems. Philadelphia, Pa.: WB Saunders Company; 1994:103-112. 42. Janisse DJ. Prescription insoles and footwear. Clin Podiatr Med Surg. 1995;12:41-61. 43. Pedorthic Footwear Association. Introduction to Pedorthics. Columbia, Md.; 1998. 44. Cohen MM, Brietstein RJ, Brill L. Wound Care Q&A: improve your treatment of Charcot foot, part II. Podiatry Today Magazine. 2000;Jul/Aug:79-81. 45. Bower AC, Allman RM. Pathogenesis of the neuropathic joint: neurotraumatic vs. neurovascular. Radiology. 1981:139-349. 46. Sanders LJ, Frykberg RG. Diabetic neuropathic osteoarthropathy: Charcot foot. In: Frykberg RG, ed. The High Risk Foot in Diabetes Mellitus. New York, NY: Churchill Livingston; 1991:297-338. 47. Greene DA, Feldman EL, Stevens M. Neuropathy in the diabetic foot: new concepts in etiology and treatment. In: Levin ME, O?Neil LW, Bowker JH, eds. The Diabetic Foot, 5th ed. St. Louis, Mo.: Mosby Year Book; 1993:135. 48. Eichenholtz SN. Charcot?s Joints. Springfield, Ill.: Charles C. Thomas; 1966. 49. Johnson JE. Surgical reconstruction of the diabetic Charcot foot and ankle. Foot and Ankle Clinics. 1997;2(1):37-55. 50. Johnson JE, O?Brien TS, Hart TS, et al. Reconstruction of the Charcot?s foot and ankle: an outcome study of long-term results. Abstract presented at the American Orthopedic Foot and Ankle Society 12th Annual Summer Meeting; June 27-30, 1996; Hilton Head, SC. 51. Banks AS. A clinical guide to Charcot foot, In: Kominsky SJ, ed. Medical and Surgical Management of the Diabetic Foot. Baltimore, Md.: Mosby; 1994:115-143. 52. Krause JO, Brodsky JW. The natural history of type 1 midfoot neuropathic feet. Foot and Ankle Clinics. 1997;2(1):1-22. 53. Frykberg RG, Mendeszoom ER. Charcot arthropathy: pathogenesis and management. Wounds. 2000;12(6 Suppl B):35B-42B. 54. Waters RL, Perry J, Antonelli D, et al. Energy cost of walking of amputees: the influence of length of amputation. Am J Bone Joint Surg. 1976;58A:42-51. 55. Marks RM. Mid-foot/mid-tarsus amputations. Amputations. Foot and Ankle. 1999;4(1):1-16. 56. Campbell JT. Syme?s, Boyd?s, Chopart?s and Pirogoff?s amputations. Amputations. Foot and Ankle. 1999;4(1): 39-62. 57. Lehman JF, Price R, Koon G. Worth the weight: prosthetic mass and gait. Biomechanics. 1998;12:15-20. Additional Resources Valmassy RL. Clinical Biomechanics of the Lower Extremities. St. Louis, Mo.: Mosby Yearbook; 1996:365-366. Armstrong DG, Abu-Rumman PL, Nixon, BP, Boulton, AJM. Continuous activity monitoring with persons at high risk for diabetis-related extremity amputation. J Amer Pod Med Assoc. 2001;91(9):451-455.